Digital Dental X-ray Unit with 4400mAh Strong Battery
If you have your own design, please contact us, there will have a professional customer service to communicate with you.
Don't have your files right now?No worries, Check out our art tips for more information.
If over 500pcs, mass production will start until pre-production sample has been approved.
Est. 7 days production
Tube voltage: 60KV
Tube current: 1.2mA
Exposure time: 0.1--2.0s
Rate power: 60VA
Distance from skin to cone: 140mm
Tube focus: 0.4mm*0.4mm
Battery: DC14.8V 4400mAh
Charger input voltage: AC100V-240V±10%
Output voltage: DC16.8V
Leak radiation: 1 Meters≤0.002mGy/h
Ambient temperature: 5℃~40℃
Relative temperature: <75%RH
Net weight: 2.6KG
Gross weight: 5.0KG
Package size(mm): 280*270*250
Small size, light weight, portable.
Memory function, just once exposure, can memory exposure parameters forever.
LCD screen, all function could see clearly.
Tact switches, smart control, convenience operating.
High Frequency power supply, low radiation, handsome appearance.
Developing in light room, image in one minute, be used for diagnoses at once.
Be use of the intraoral cavity digital imaging system.
Characteristics and Main Use
This portable dental x-ray unit is high frequency machine. It has excellent image quality, portable storage, save more space. It uses high frequency and DC international power supply. All of the components installed in the central PC board concentrated. Shock, set-up, electron tubes, all of them are insulation vacuum, sealed stereotype protection. There are also manual buttons installed on the surface of shell, as well as batteries and charges. This unit mainly suitable for oral pre-treatment for learning internal organizational structure, root depth and so on, is indispensable in daily life clinic equipment, especially for dental implant surgery. It also can connect with sensor, is great convenience.
Checking and Security
*Caution: Please learn relative security measures and functions given in the file or on the unit.
*Using the products after checking battery and the location of the device, make sure that is operates properly.
*Checking all the power cords whether correctly and connected formally.
*Operating the device in the designated order.
*Recheck the cone connect with the patient’s face closely.
*Keep the device and the cone clean.
*Please use or operate the device and parts after taking a necessary periodic inspection.
*If you discover some difference in the sensor images even if you inspected its x-ray exposure time in the same way as before, then it means that the radiation of x-ray is abnormal, you must stop using it immediately and allow it to be inspected.
*Don’t use or store near the place which is possible to fire.
*Don’t use or store where air pressure or temperature or moisture exceeds range.
*Keep good ventilation and avoid direct sunlight, protect against the eroding of dust and corrosive air.
*Don’t use or store near to chemicals or combustible gas.