Dental X Ray Unit PORX Wireless Handheld Portable System
If you have your own design, please contact us, there will have a professional customer service to communicate with you.
Don't have your files right now?No worries, Check out our art tips for more information.
If over 500pcs, mass production will start until pre-production sample has been approved.
Est. 7 days production
Rechargeable Lithium Polymer battery: 24V
Maximum battery charge: 25.2V
Battery current: 2mA, 60KV
Battery Output: 10A
Power supply: JBL7451250750100FJ, Bridge Power Corp.
Input: AC100-240V, 50-60HZ 1.0A
Output: DC25.2V, 0.75A
Cable length of DC output: 1550mm
Cable length of AC input: 1550mm
Cable length of remote controller: 3000mm
X-Ray control and Generator
Exposure time range: 0.01-1.6sec
Maximum duty cycle: 1:60
Maximum inherent filtration: 1.6mm Al
Maximum output power: 120W nominal at 60KV, 2mA
Generator rating: 2mA at 60KV±5%
Leakage technique factors: 60KV, 2mA, 1.6sec
Minimum source to skin distance: 10cm, 20cm (long cone)
X-ray field size and configuration: 6cm diameter circle
The unit is particularly provided for synthesis hospitals,cavity specialty hospital and dental clinic to take photo.
Environmental and working conditions
a. Condition of transportation and store
Environmental temperature: 10~40°C
Relative humidity: 5%-90% RH
b. Conditions of operation
Environmental temperature: 10~30°C
Relative humidity: 10%-85% RH
MDD(93/42/EEC): AnnexIX [Class II b]
IPX specification [IPO: do not operate under wet conditions]
Type of protection against electric shock [Internal powered device]
Degree of protection against electric shock [Type B applied part]
For use in environments where no flammable anesthetics and/or flammable.
Cleaning agents are present;non-alcohol based disinfectant only-wipes or cloth.
Dampened with liquid/spray.